- Commodity Quality Certification: 3C
- Material: Metal
- Brand Name: HARBOR DENT
- Origin: Mainland China
Two or Four-holes handpiece tubing*2 units
Φ8gas pipe (1.6m)*2 units
Φ12drain hose (1.2m)*1 unit
Pneumatic weak suction apparatus *1 unit
Circular pedal switch*1 unit
Triple gun*1 unit
600ml Water tank*1unit
Button Function Description
1.Pressure Gauge: Display pressure level while high/low speed handpiece is during operation
2.Gas Switch: Adjust pressure level while high/low speed handpiece is during operation.
3.Nebulizer Switch: Adjust nebulizer pressure while high/ow speed handpiece requires nebulization function.
4. Gas Switch: Adjust pressure level while high/low speed handpiece is during operation.
5. Nebulizer Switch: Adjust nebulizer pressure while high/low speed handpiece requires nebulization function.
6.Water Switch: Switch on when high/low speed handpiece requires water. NO NEED TO TURN OFF
7.Water Tank Switch: When high/low speed handpiece requires water, switch on the water tank first.
8.Tap Water Switch: When high/low speed handpiece requires tap water, please switch on this button and switch off the water tank.
9.Water Filter: filter inpurities in the water, screw it down when water is full, empty thewater bucket, and screw it up.
10.Suction Water Joint: Connect 610soft tubing, drain away the waste water, please prepare yourOWN waster water bucket.
11.Tap Water Joint: Connect this joint when tap water is required
12.Intake Joint Please connect with oilless air compressor.
13. Water Tank: Storage pure water. (we suggest you to use pure water instead of tap water,which is better for the lifetime of this device.)
14. Height adjusting rotary button: Able to adjust its height to 10cm.
Dental curing light
3.1 Three parts of the package in one unit of device
① Four wheels at the end of basin:
② Four screws on the connecting square tubing in the middle;
③ One ratory button to adjust height is packed with tool basin.
3.2 Honeycomb carton package
3.3 Net Wight: 16KG
3.4 Gross Weight: 23KG
3.5 Package Dimension: 690*520*410 (mm)
Warranty-period, following condition are NOT in the warranty scope:
A. Man-made damage
B. Damages caused by abnormal operation.
C. Darmages caused by operation that NOT according to user